Nanopore sequencing offers advantages in all areas of research. Our offering includes DNA sequencing, as well as RNA and gene expression analysis and future technology for analysing proteins.
You can contact our Customer Services and Technical Services teams round the clock using the contact support function.
Please try the following steps before repeating the platform QC:
1. Remove the flow cell from the MinION and let it cool for a few minutes
2. Unplug the MinION from the computer and let it cool for a few minutes
3. Reboot the computer
4. Insert the flow cell, ensuring you exert moderate downwards pressure so that the ASIC chip
5. Relaunch the software and plug the MinION into the computer
6. Repeat the Platform QC/flow cell check
If the pore number on the flow cell equals 0, then please do the following as this could result from the presence of a bubble near the electrode, located in the inlet channel, preventing current flow to the pore array area.
1. Remove 15-20 uL of the buffer from the priming port (with the priming port completely open and the Spot-On port covered). There is enough buffer in the outlet channel so that even if you remove about 20 uL, the flow cell will still be covered with liquid at all times.
2. Repeat the Platform QC to see if you obtain different results
Please note that exposure to air or freezing will irreversibly damage the pores. Please ensure the pore area is always covered with buffer during flow cell handling and the storage is within the 2-8°C range (storage instructions here).
Don’t worry when you check your flow cell if it looks like there is liquid condensation inside the packaging – this is purely a cosmetic issue and does not affect flow cell performance.
Our packaging contains moisture control technology that ensures the flow cells are environmentally controlled during shipping and storage.
Fully released flow cells that fail to pass QC within 12 weeks of arrival can be reported to the Support team.
Flow cells that are in Early Access or Developer Release may have different warranty periods.
Please report any flow cells that fail to pass QC within two business days of the initial flow cell check.